Department of Mechanical and Materials Engineering,University of Nebraska–Lincoln,Lincoln, NE, 68508e-mail: email@example.com
Department of Mechanical Engineering,University of Colorado at Boulder,Boulder, CO 80309-0427e-mail: firstname.lastname@example.org
Department of Surgery,University of Colorado at Denver,Aurora, CO 80045e-mail: email@example.com
Department of Mechanical Engineering,University of Colorado at Boulder,Boulder, CO 80309-0427e-mail: firstname.lastname@example.org
Manuscript received March 15, 2013; final manuscript received April 25, 2013; published online July 3, 2013. Assoc. Editor: Arthur G. Erdman.
Schematic of MFS in vivo. As described in , characterization of the MFS is achieved by relating sensor pressure, Pi, to intraluminal pressure, Pi, abdominal pressure, Pa, contact pressure, Pc, and sensor temperature, Ts. Sensor output is the contact force experienced by the MFS. A purpose of this research is to replace the latex balloons and plastic components with their biocompatible equivalents.
Human MFS mounted to the end of a grasper showing biocompatible components. The Sensor is inflated for calibration prior to deflation and insertion (top). Deflated sensor shown being inserted through a 10 mm trocar ready for placement in the small intestine (bottom).
Some tools below are only available to our subscribers or users with an online account.
Download citation file:
Customize your page view by dragging and repositioning the boxes below.
© 2015 ASME
The American Society of Mechanical Engineers
Sorry! You do not have access to this content. For assistance or
to subscribe, please contact us:
Sign in or create your free personal ASME account. This will give you the ability to save search results, receive TOC alerts, RSS feeds, and more.
Sign into or create your free personal account